Download the most recent User Manual and Quick Start Guide for your WaterLink® Spin Touch®.
Watch video tips about important procedures for your WaterLink® Spin Touch®.
Read about important procedures, warranty information, tips and troubleshooting for your WaterLink® Spin Touch®.
WATERLINK® SPIN TOUCH® DOCUMENTS
Please check back regularly for the most recent WaterLink® Spin Touch® Manual and Quick Start Guides.
WaterLink® Spin Touch® User Manual
WaterLink® Spin Touch® Quick Start Guide
VIDEO TUTORIAL FOR PROPER FILLING TECHNIQUE FOR WATERLINK® SPIN DISK
Problem: Disc Not Spinning
Reason: Lid open, meter not powered on, low battery, disk or disk cover pressed down too tightly on hub
Solution: Close lid, power on meter, charge the battery or plug meter into a stable power source, remove the disk/disk cover and place back in the meter more gently
Problem: Unexpected Results
Reason: Did not use disk cover
Solution: Place the disk cover on the disk and close the lid
Problem: Unexpected Low Chlorine Results
Reason: High chlorine concentrations will bleach chlorine, and other reagents
Solution: Check sample with chlorine test strips (Code 2987)
Problem: Trouble Connecting to Bluetooth Device
Reason: Bluetooth not enable
Solution: Enable Bluetooth
Problem: Trouble Connecting to Bluetooth Device
Reason: Too many Bluetooth devices near the meter
Solution: Have only one device near the meter
Problem: “Source” Appears on Test Screen Instead of “Sanitizer”
Reason: Wrong Market Type was chosen
Solution: Go to Settings. Change Market Type to Pool/Spa
TROUBLESHOOTING WITH THE METER CHECK DISK
Range Check Procedure
The Meter Check Disk is used to verify the performance of a meter. It is not used to recalibrate the meter. Meters are calibrated at the time of manufacture. The meter is performing satisfactorily if readings using the Meter Check Disk are within the ranges provided on the Meter Check Disk pouch.
Range specifications are specific to the disk identified by the serial number on the pouch. The range specifications will vary from disk to disk. The exact readings from a specific disk may vary from meter to meter.
If the results are not within the range shown on the pouch, contact tech service.
The Meter Check Disk is inserted in the meter and Calibration is chosen from the Settings Menu to perform an angle calibration which checks the alignment of the hub and disk. The results are analyzed and reported as Pass or Fail. If the measurements pass, the settings will be saved. If the analysis fails, contact tech support.
The optical system of the WaterLink® Spin Touch® must be kept clean and dry for optimal performance. Dry the disk with a lint-free wipe before placing it into the chamber to avoid introducing moisture. For best results, store the instrument in an area that is dry and free from aggressive chemical vapors. Clean the exterior housing with a damp, lint-free cloth. Do not allow water to enter the light chamber or any other parts of the meter. To clean the light chamber and optic lenses, point a can of compressed air into the light chamber and the lid and blow the pressurized air into the light chamber and lid. Focus the pressurized air around the LEDs which are the small round lenses positioned at 2:00, 4:00, 6:00, 8:00, 10:00 and 12:00 in the lid. The photodiodes are located on the bottom of the chamber around the hub. This area must be kept clean and dry. Use a cotton swab dampened with Windex® window cleaner to gently swab the LED and photodiode lenses. Do not use alcohol; it will leave a thin residue over the optics when dry.
Remove smudges due to routine use from the touchscreen with the Cloth Wipe (Code 3580-WIPE). Use a cloth dampened with alcohol for more thorough cleaning when necessary. Do not use Windex® window cleaner, or similar cleaners, on the touchscreen.
Should it be necessary to return the meter for repair or servicing, pack the meter carefully in a suitable container with adequate packing material. A return authorization number must be obtained from LaMotte Company by calling 800- 344-3100, ext. 2 (US only) or 410-778-3100, ext. 2, faxing 410-778-6394, or emailing firstname.lastname@example.org. Often a problem can be resolved over the phone or by email. If a return of the meter is necessary, attach a letter with the return authorization number, meter serial number, a brief description of problem and contact information including phone and FAX numbers to the shipping carton. This information will enable the service department to make the required repairs more efficiently.
Waste Electrical and Electronic Equipment (WEEE)
Natural resources were used in the production of this equipment. This equipment may contain materials that are hazardous to health and the environment. To avoid harm to the environment and natural resources, the use of appropriate take-back systems is recommended. The crossed out wheeled symbol on the meter encourages the use of these systems when disposing of this equipment.
Take-back systems will allow the materials to be reused or recycled in a way that will not harm the environment. For more information on approved collection, reuse, and recycling systems contact local or regional waste administration or recycling services. Do not incinerate the equipment.
The disks cannot be reused. Over time, the water in reacted disks will evaporate. Disks can be recycled. Warning: Recyclers should check with the local authorities. Some states may require that no chemical residue remains on the plastic or may not be able to accept plastic waste with stainless steel mixing beads. Used disks may be returned, at the customer’s expense, to LaMotte for recycling.
Disk Recycling Program
LaMotte Company is committed to protecting our environment and preserving our natural resources. As part of our commitment, customers can return used Spin disks to LaMotte Company for recycling. The collected disks will be sent to a certified recycling center where they will recover and reuse both the metal and plastic components. While the recycling is free, customers are responsible for the cost to send the disks to LaMotte Company.
How to Participate
For question, please contact LaMotte Technical Service: Phone: 1-800-344-3100 x 2; E-mail: email@example.com
Packaging and Returns
Experienced packaging personnel at LaMotte Company assure adequate protection against normal hazards encountered in transportation of shipments. After the product leaves the manufacturer, all responsibility for its safe delivery is assured by the transportation company. Damage claims must be filed immediately with the transportation company to receive compensation for damaged goods. Should it be necessary to return the instrument for repair or servicing, pack the instrument carefully in a suitable container with adequate packing material. A return authorization number must be obtained from LaMotte Company by calling 1-800-344-3100 or 1-410-778-3100, ext. 2 or emailing tech@lamotte. com. Attach a letter with the authorization number to the shipping carton which describes the kind of trouble experienced. This valuable information will enable the service department to make the required repairs more efficiently.
Read the instruction manual before attempting to set up or use the instrument. Failure to do so could result in personal injury or damage to the meter. The WaterLink® Spin Touch® should not be stored or used in a damp or excessively corrosive environment. Care should be taken to prevent water or reagents from entering the photometer chamber. Wet disks should never be put into the photometer chamber.
Read the safety precautions on the labels of all reagent containers and packaging prior to use. Safety Data Sheets (SDS) can be found at www.lamotte.com. Additional emergency information for all LaMotte reagents is available 24 hours a day from the National Poison Control Center at 1-800-222-1222 or by contacting the 24 hour emergency line for ChemTel at 1-800-255-3924 (USA, Canada, Puerto Rico). For locations outside of the North American continent call 813-248-0585 collect.
Ensure that the protection provided by this equipment is not impaired. Do not install or use this equipment in a manner that is not indicated in this manual.
Limits of Liability
Under no circumstances shall LaMotte Company be liable for loss of life, property, profits, or other damages incurred through the use or misuse of its products.
LaMotte Company warrants this instrument to be free of defects in parts and workmanship for 2 years from the date of shipment. If it should become necessary to return the instrument for service during or beyond the warranty period, contact our Technical Service Department at 1-800-344-3100 or 1-410-778-3100, ext. 2 or firstname.lastname@example.org for a return authorization number or visit www.lamotte.com for troubleshooting help. The sender is responsible for shipping charges, freight, insurance and proper packaging to prevent damage in transit. This warranty does not apply to defects resulting from action of the user such as misuse, improper wiring, operation outside of specification, improper maintenance or repair, or unauthorized modification. LaMotte Company specifically disclaims any implied warranties or merchantability or fitness for a specific purpose and will not be liable for any direct, indirect, incidental or consequential damages. LaMotte Company’s total liability is limited to repair or replacement of the product. The warranty set forth above is inclusive and no other warranty, whether written or oral, is expressed or implied.
To register your meter with the LaMotte Service Department, go to www.lamotte.com and choose SUPPORT on the top navigation bar.
The WaterLink® Spin Touch® is compatible with iPhone 6 Plus, iPhone 6, iPhone 5s, iPhone 5c, iPhone 5, iPad Air 2, iPad Air, iPad mini 3, iPad mini 2, iPad mini, and iPad (4th generation).
Compatible with Android devices.
WaterLink® and DataMate® are registered trademarks of LaMotte Company.
Disk Patent No. 8,734,734; FCl Patent No. 8,987,000; TCl Patent No. 8,993,337; FCl EU Patent No. EP2784503 A1